Please fill out the following form to submit a Request
for Quote to any of the following companies listed on
Get Your Company Listed on this Power Page
This article is a complete guide to differential pressure switches and their use.
You will learn:
What is Differential Pressure?
How Differential Pressure is Measured
Types of Differential Pressure Switches
Reasons for Measuring Differential Pressure for Switch Applications
Applications of Differential Pressure Measurements
And so much more …
Chapter One: What is Differential Pressure?
Differential pressure is the difference in pressure between two points that is measured using different types of pressure sensing devices. An essential part of differential pressure measurement are gauges and switches that are used to control and monitor pressure. When activated, switches sound an alarm or break a circuit when pressure rises or falls beyond preset values.
The three types of pressure are relative pressure, absolute pressure, and differential pressure. Relative pressure measures the difference between absolute pressure and another pressure value while absolute pressure is the pressure relative to zero pressure in a vacuum or empty space.
The complexity of differential pressure is due to it measuring the pressure between two variables with unknown values. The pressure is measured by sensors that are connected to the two sources. Gauges or switches calculate the difference in pressure between the sources. When pressure reaches a certain point, a switch will make or break a circuit. The need to monitor differences in pressure is dependent on applications that require the use of fluids and gases.
The choice of the method for measuring differential pressure depends on the application, the pressure range, maximum pressure for the application, and the substance that is being measured. It is possible for a technician to measure differential pressure and work out the difference. Inaccuracy and the time involved to make calculations is the reason that DP gauges and switches have been invented.
Chapter Two: Measuring Differential Pressure for Liquids
Measuring differential pressure for liquids involves measuring the pressure drop across a restriction in the flow of the liquid. The pressure drop is proportional to the flow rate of the liquid. Measuring differential pressure in liquid systems requires specialized equipment and careful selection of materials.
The choice of equipment and material depends on the specific requirements of the system, including the type of liquid being measured, the temperature and pressure of the system, and the desired level of accuracy. Commonly used methods include using manometers, pressure gauges, and differential pressure transmitters.
One of the most important factors in measuring differential pressure for liquids is selecting the correct material for the equipment. The material must be compatible with the liquid being measured, and it must be able to withstand the temperature and pressure of the system. Common materials used for measuring differential pressure in liquid systems include stainless steel, brass, and plastic.
Stainless steel is often preferred because it is a durable and corrosion-resistant material. It is able to withstand harsh and corrosive environments, making it ideal for applications such as chemical processing and oil and gas production. It also has a high melting point and is able to withstand high temperatures, which makes it suitable for high-temperature applications.
Brass, meanwhile, is preferred because it is a relatively low-cost material that is easy to machine. It is commonly used for lower-pressure applications and is often used in applications where it is necessary to have a tight seal between components.
Plastic, on the other hand, is often chosen because it is lightweight and resistant to many chemicals. It is also a cost-effective alternative to metal materials. Plastic materials such as PVC, polypropylene, and PTFE (polytetrafluoroethylene) are commonly used for applications that require resistance to harsh chemicals and high temperatures.
Chapter Three: Measuring Differential Pressure for Gases
A common method for measuring differential pressure in gases is the use of differential pressure transmitters that use a diaphragm to sense the difference in pressure between two points and convert the difference into an electrical signal that can be read by a controller or computer.
Orifice plates, Venturi tubes, and flow nozzles are used for measuring differential pressure in liquids. When measuring gasses, the temperature and pressure of the gas has to be considered since they affect the accuracy of measurements.
Gas leaks that can occur in piping or equipment used to transport the gas and affect the results in differential pressure measurements of gases. Equipment for differential pressure gas measurements are designed for use with gases and have safety protocols outlined for the process. Measuring differential pressure for gases is a process used to ensure the safety and efficiency of an application.
Chapter Four: How Differential Pressure Switches Work
Differential pressure switches have three basic components: sensing element, stable or range spring, and snap acting microswitch. These three parts are combined in various combinations for the manufacture of several kinds, types, and designs of differential pressure switches for monitoring industrial fluid and gas systems.
A differential pressure switch is a simple electromechanical switch that works on the basic ideas of levers and opposing forces. They are designed to sense the difference between pressure at two different points or sources, which are connected to a sensing diaphragm. When a difference of pressure is created, force is generated that moves a tension spring that moves a connecting rod that produces sufficient movement to activate a microswitch.
The design of differential pressure switches includes two ports, one for high pressure and one for low pressure, that are separated by a diaphragm. Differences in pressure between the two ports causes axial movement, measuring travel, of the diaphragm against the measuring spring that moves the connecting rod to the microswitch.
The sensing of pressure involves pressure created by a media, such as gas or liquid, that causes physical displacement to move the pointer on a scale, an electrical measurable response, or capacitance change. The types of sensing elements include diaphragms, capsules, Bourdon tubes, and bellows that convert pressure to displacement.
Diaphragm - A pressure sensing diaphragm is a circular plate that is fixed around its edges and exposed to the pressure media. On the opposite side of the diaphragm may be a sealed chamber or gauge. When pressure is applied, the diaphragm deflects it proportional to the amount of force applied.
Capsule - A capsule has two diaphragms where pressure acts on both diaphragms, which allow for the measurement of low pressure that cannot be measured by a single diaphragm.
Bourbon Tube - Bourbon tubes can be C shaped or helical with an oval cross section. When pressure is applied, it changes the oval to a circle, which distorts the tube and moves an indicator needle.
Bellows - A bellows expands in response to pressure and moves a pointer to indicate the level of pressure.
The movement of the sensing element is resisted by the spring, which determines the differential pressure range in accordance with the movement of the sensing element that triggers the electrical switch. The actuation point of the switch is set by adjusting the spring’s tension or compression.
The microswitch turns the differential pressure switch on or off. The difference in pressure is converted to axial movement that is sent to the microswitch that has electrical contacts that actuate depending on switch points or setpoints.
Chapter Five: Differential Pressure Measurement Units
Differential pressure is measured in Pascals (Pa), pounds per square inch (psi), inches of water column (inWC), and millimeters of mercury (mmHg). The choice of units depends on the specific application and the preference of the user. The two most common units for differential pressure measurement are Pascals and pounds per square inch.
A Pascal is a unit of pressure in the International System of Units (SI), which is the system of measurement used in most countries. It is defined as the pressure exerted by a force of one Newton per square meter. In the context of differential pressure, a Pascal represents the pressure difference between two points in a fluid system or a gas system.
Pounds per square inch (psi) are units of pressure used in the United States and other countries. It is defined as the pressure exerted by a force of one pound-force applied to an area of one square inch. In the context of differential pressure, a psi represents the pressure difference between two points in a fluid system or a gas system.
There are several differences between psi and Pascals. A major difference is that Pascals are generally used for scientific and engineering calculations, while psi is more common in industrial applications. Another difference is that psi is often used in the United States, while Pascals are more common in other parts of the world.
Understanding the differences between the measurement units used to measure differential pressure is important for several reasons. To begin with, it is important to ensure that the measurement unit used is appropriate for the application. For example, if a pressure sensor is calibrated in pounds per square inch (psi) and the application requires a reading in Pascals (Pa), the conversion factor must be applied correctly to ensure that the reading is accurate.
Using the wrong measurement unit can lead to errors in measurements and inaccurate readings. This can be problematic in critical applications such as medical or aerospace where accuracy is essential. Understanding the differences between the measurement units helps ensure that the appropriate equipment is used for the application. For example, equipment designed to measure pressure in pounds per square inch (psi) may not be suitable for an application that requires measurements in Pascals (Pa).
Chapter Six: Equipment Used for Measuring Differential Pressure
Although differential pressure switches are a widely used method for monitoring and controlling differential pressure, there are several other types of devices used,which include manometers, pressure gauges, differential pressure transmitters, orifice plates, Venturi tubes, flow nozzles, pilot tubes, thermal flow meters, and Coriolis flow meters. Each type has design features that make them suitable for specific applications or purposes.
A manometer is a simple device that measures pressure by comparing it to the pressure of a fluid in a column. There are several different types of manometers, including U-tube, inclined-tube, and well-type manometers. Manometers are often used in HVAC systems, as well as in laboratory and industrial settings.
A pressure gauge is a device that measures the pressure of a fluid or gas. There are several different types of pressure gauges, including Bourdon tube gauges, diaphragm gauges, and capsule gauges. Pressure gauges are commonly used in industrial and commercial settings, and are often used in conjunction with differential pressure transmitters.
Differential Pressure Transmitters
Differential pressure transmitters are devices that measure the difference in pressure between two points in a system. These devices are used in HVAC systems, as well as in industrial and commercial settings. They can be used in conjunction with other measurement devices, such as orifice plates, Venturi tubes, and flow nozzles.
An orifice plate is a thin metal plate with a small hole in the center. When placed in a pipe, the orifice plate creates a pressure drop across the plate, which can be measured to determine the flow rate of the fluid or gas. Orifice plates are commonly used in HVAC systems, as well as in industrial and commercial settings.
A Venturi tube is a device that uses a narrowing of a pipe to create a pressure drop, which can be measured to determine the flow rate of the fluid or gas. Venturi tubes are commonly used in HVAC systems, as well as in industrial and commercial settings.
A flow nozzle is a device that uses a constriction in a pipe to create a pressure drop, which can be measured to determine the flow rate of the fluid or gas. Flow nozzles are commonly used in HVAC systems, as well as in industrial and commercial settings.
A pitot tube is a device that measures the pressure of a fluid or gas at a single point. Pitot tubes are commonly used in aviation to measure airspeed, as well as in HVAC systems and industrial and commercial settings.
Thermal Flow Meters
A thermal flow meter is a device that measures the flow rate of a fluid or gas by measuring the temperature difference between two points. These devices are commonly used in industrial and commercial settings.
Coriolis Flow Meters
A Coriolis flow meter is a device that measures the flow rate of a fluid by measuring the twisting or bending of the fluid as it flows through a pipe. Coriolis flow meters are commonly used in industrial and commercial settings.
Leading Manufacturers and Suppliers
Chapter Seven: Calibration and Maintenance of Differential Pressure Equipment
Calibration is a critical aspect of maintaining the accuracy and reliability of differential pressure measurement equipment. It is used to verify the accuracy of a device's measurements and adjust it as necessary to ensure it is within an acceptable range of error. It is recommended to calibrate differential pressure measurement equipment annually, or more frequently depending on the criticality of the application, to ensure that they are operating correctly and provide accurate measurements.
The process for calibration is performed using a reference standard or calibration equipment that is traceable to a national or international standard. The equipment must be properly maintained and calibrated to ensure its accuracy. Calibration laboratories or service providers should have a quality management system in place and should be accredited to international standards, such as ISO/IEC 17025.
Calibration equipment used for differential pressure measurement include deadweight testers, pressure calibrators, digital pressure gauges, or reference pressure transmitters. They provide known pressure references that can be used to compare against the readings obtained from the differential pressure measurement equipment being calibrated. The types of calibration equipment used should be suitable for the range and accuracy required for the equipment being calibrated. Proper handling and storage of calibration equipment are also critical to maintain their accuracy.
Regular maintenance of differential pressure measurement equipment is also essential to ensure its continued accuracy and reliability. This may include routine cleaning, inspection, and replacement of worn or damaged parts. Regular maintenance can help to identify issues before they become major problems and ensure that the equipment is functioning optimally.
Chapter Eight: Applications of Differential Pressure Measurements
Differential pressure measurements have numerous applications across many industries. They go beyond traditional and regular pressure measurements and are used to check the condition of filters, liquid levels, flow rates, and torque output. Differential pressure measurements help determine the desired pressure level for an application and the maximum pressure at which it can be completed.
One of the most common applications of differential pressure is in heating, ventilation, and air conditioning (HVAC) systems. Differential pressure measurements are used to ensure that air flows properly through ductwork, filters, and other components of the HVAC system. They can also be used to measure the efficiency of the system and identify any blockages or leaks that may be affecting performance.
Differential pressure measurements are essential in chemical manufacturing processes, where they are used to monitor pressure drops across filters, pumps, and other equipment. This helps to ensure that the process is running efficiently and that there are no blockages or leaks that could affect the quality of the final product.
Differential pressure measurements are widely used in the medical industry, particularly in hospitals and laboratories. They are used to maintain a clean and sterile environment, such as in operating rooms, isolation wards, and research facilities. Differential pressure sensors are used to monitor the pressure difference between two rooms or areas, ensuring that air flows from clean areas to contaminated areas, minimizing the risk of infection.
In the environmental industry, differential pressure measurements are used to monitor air quality in buildings and industrial facilities. They can also be used to measure the flow of gasses and liquids through pipelines and other systems. For example, differential pressure sensors are used to monitor the pressure drop across filters and membranes, ensuring that they are operating efficiently.
Differential pressure measurements are used extensively in the aerospace industry, particularly in aircraft and spacecraft. Differential pressure sensors are used to measure the airspeed of the aircraft and to monitor the pressure difference between the cabin and the outside environment. They can also be used to monitor the fuel and hydraulic systems, ensuring that they are functioning properly.
In the automotive industry, differential pressure measurements are used to monitor the performance of engines and other systems. They can be used to measure the pressure drop across air filters, oil filters, and fuel filters, ensuring that they are clean and functioning properly. Differential pressure sensors can also be used to monitor tire pressure, which is important for safety and fuel efficiency.
Differential pressure measurements are used in the food industry to monitor and control the production process. They can be used to measure the pressure drop across filters and other equipment, ensuring that the food is safe and free from contaminants. Differential pressure sensors are also used to monitor the pressure difference between different areas of the production facility, ensuring that air flows properly and that there are no cross-contamination issues.
Water Treatment Industry
In the water treatment industry, differential pressure measurements are used to monitor the pressure drop across filters, membranes, and other equipment. They can be used to ensure that the system is operating efficiently and that the water is safe and free from contaminants. Differential pressure sensors are also used to monitor the pressure difference between different areas of the treatment plant, ensuring that air flows properly and that there are no cross-contamination issues.
Differential pressure measurements are essential in the pharmaceutical industry, where they are used to maintain a clean and sterile environment in production facilities. Differential pressure sensors are used to monitor the pressure difference between different areas of the facility, ensuring that air flows properly and that there are no cross-contamination issues. They are also used to monitor the pressure drop across filters and other equipment, ensuring that the production process is running efficiently and that there are no blockages or leaks.
Power Generation Industry
Differential pressure measurements are used in the power generation industry to monitor the performance of turbines, boilers, and other equipment. They can be used to measure the pressure drop across filters, pumps, and other components, ensuring that they are functioning properly and efficiently. Differential pressure sensors can also be used to monitor the pressure difference between different areas of the power plant, ensuring that there are no leaks or other issues that could affect performance.
Pneumatic Control Systems
Differential pressure measurements are used in pneumatic control systems to monitor and control the flow of compressed air. Differential pressure sensors are used to measure the pressure difference between two points in the system, allowing engineers to adjust the flow rate and pressure as needed.
Chapter Nine: Benefits of Differential Pressure Measurements and Equipment
Differential pressure measurements and equipment are used to improve efficiency, safety, and accuracy. They are capable of measuring minute changes in pressure using a wide range of pressure settings and determining the maximum pressure an application can withstand. Differential pressure measurement equipment has to be custom designed to fit the needs of an application with changes in port positioning, the addition of ports, and different process connections.
Differential pressure measurements can help identify areas of inefficiency in a system. By measuring the pressure drop across filters, pumps, and other components, engineers can identify blockages or leaks that can impact the overall efficiency of the system.
Differential pressure measurements can be used to monitor pressure differences between different areas of a system or facility, ensuring that air flows properly and that there are no cross-contamination issues. In medical and research facilities, for example, differential pressure is used to create a clean and sterile environment that is essential for patient safety and laboratory experiments.
Differential pressure measurements provide accurate readings of pressure differences, allowing engineers to precisely control the flow of gasses or liquids in a system. This is important in applications such as HVAC systems or chemical manufacturing, where precise control of pressure and flow rates is critical.
Differential pressure measurements can be used in a wide range of applications and industries, from aerospace to water treatment. The versatility of differential pressure measurement technology makes it an essential tool for many different types of systems.
Differential pressure measurements can lead to cost savings in several ways. By measuring the pressure drop across filters, pumps, and other components in a system, engineers can identify blockages or leaks that can impact the overall efficiency of the system. This allows them to identify areas of inefficiency and take corrective action to reduce energy consumption and maintenance costs. For example, if a filter is clogged, it will create a pressure drop that can reduce airflow and increase energy consumption. By measuring the pressure drop across the filter, engineers can identify the problem and replace the filter, which can lead to significant cost savings in the form of reduced energy consumption and maintenance costs.
In addition to these general benefits relating to differential pressure measurements there are a few benefits relating specifically to differential pressure measurement equipment which may be of interest, including:
Differential pressure equipment is typically designed to withstand harsh environments and extreme temperatures. This makes it an ideal tool for use in applications such as mining or oil and gas production, where equipment must be able to withstand harsh conditions.
Ease of Use
Differential pressure equipment is generally easy to use and requires minimal training. This makes it an ideal tool for use in a wide range of industries and applications.
Differential pressure equipment is typically designed to be highly reliable and durable, providing accurate measurements over an extended period of time. This reduces maintenance costs and ensures that the equipment is always ready to use when needed.
Occupational Safety and Health Administration (OSHA) Regulations
OSHA has several regulations related to pressure vessels, including regulations related to the design and construction of pressure vessels, as well as requirements for testing and inspection of pressure vessels.
National Fire Protection Association (NFPA) Standards
The NFPA has several standards related to pressure vessels and pressure relief devices, which may impact the design, construction, and use of differential pressure equipment.
American Society of Mechanical Engineers (ASME) Standards
The ASME has several standards related to pressure vessels and pressure relief devices, which are widely used in the design and construction of differential pressure equipment.
International Organization for Standardization (ISO) Standards
The ISO has several standards related to pressure measurement, including standards related to differential pressure measurement.
National Institute of Standards and Technology (NIST)
NIST is a U.S. government agency that is responsible for developing and maintaining measurement standards. NIST provides calibration services for differential pressure equipment and maintains the primary standards for pressure measurement.
The American Industrial Hygiene Association (AIHA)
The AIHA is a professional organization that provides guidance on occupational health and safety, including guidance on the use and interpretation of differential pressure measurements in industrial settings.
International regulatory agencies include the International Electrotechnical Commission (IEC), the International Society of Automation (ISA), and the European Pressure Equipment Directive (PED), among others. The standards for these organizations are focused on ensuring the safety and reliability of pressure equipment, and the impact of design, construction, and their use. Pressure differential gauge and switch manufacturers work in unison with the regulatory organizations and governments to ensure the compliance of their products.
Chapter Eleven: Considerations Regarding Differential Pressure Measurements and Equipment
While differential pressure is an important measurement, it has some considerations to take into account. Each of the various problems can affect the accuracy and response time of the sensing system. Differential pressure measurements are another tool that is of use in measuring and monitoring manufacturing processes. In essence, they are one of many tools that work in combination to supply data and readings and should be treated as such.
Differential pressure measurements can be affected by factors such as temperature, humidity, and atmospheric pressure, which can affect the accuracy of the measurements. Furthermore, some types of differential pressure sensors can experience drift over time, which can also impact the accuracy of the measurements.
Limitations in Low-Pressure Measurements
Some differential pressure sensors may have difficulty accurately measuring very low pressures, which can limit their use in certain applications.
Limitations in High-Pressure Measurements
Similarly, some differential pressure sensors may have difficulty accurately measuring very high pressures, which can limit their use in certain applications.
Differential pressure sensors require regular maintenance and calibration to ensure accurate measurements over time. This can be time-consuming and costly, particularly for large industrial systems.
Differential pressure sensors can be expensive, particularly if they require specialized equipment or calibration procedures. This can make them prohibitive for some applications, particularly in smaller facilities.
Differential pressure measurements may not be suitable for all applications. For example, some systems may require more precise pressure measurements than differential pressure sensors can provide, or may require measurements of other parameters such as flow rate or temperature.
Chapter Twelve: The Future of Differential Pressure and Equipment
Regarding differential pressure equipment for the future, one area of interest is the integration of differential pressure sensors with other advanced technologies such as Artificial Intelligence (AI), Machine Learning (ML), and the Internet of Things (IoT). Combining these technologies makes it possible to create more advanced and intelligent differential pressure equipment that can adapt to changing conditions, predict failures before they occur, and optimize performance in real-time. This will have significant benefits for industries such as manufacturing, energy, and healthcare.
Differential pressure sensors and transmitters are commonly used to measure pressure drops across components such as filters, heat exchangers, and valves. By monitoring these pressure drops, engineers can optimize equipment performance, reduce energy consumption, and minimize downtime.
Differential pressure sensors are often used in process control systems to monitor flow rates, pressure, and level. With the rise of Industry 4.0 and the Internet of Things, there is likely to be an increased demand for intelligent sensors that can provide real-time data and support automated decision-making.
Differential pressure sensors are used in many environmental applications, such as monitoring air flow in HVAC systems, measuring gas emissions from industrial processes, and detecting leaks in pipelines. As concerns about climate change and pollution continue to grow, the demand for accurate and reliable environmental monitoring is likely to increase.
Differential pressure sensors are commonly used in medical equipment such as ventilators, anesthesia machines, and blood pressure monitors. As the global population ages and healthcare technology continues to advance, there is likely to be a growing demand for high-quality medical equipment that relies on differential pressure technology.
Safety and Security
Differential pressure sensors are used in many safety and security applications, such as monitoring air pressure in clean rooms, detecting gas leaks in industrial facilities, and maintaining pressure in aircraft cabins. As concerns about safety and security continue to grow, the demand for reliable and accurate differential pressure sensors is likely to increase.
Differential pressure sensors are used to monitor water levels and pressure in water treatment plants, distribution networks, and irrigation systems. As the world's population continues to grow, and water resources become increasingly scarce, the demand for efficient and reliable water management systems is likely to increase.
Differential pressure sensors are used to measure fluid flow rates in many renewable energy systems, such as wind turbines and hydropower plants. As the world shifts towards renewable energy sources, the demand for efficient and accurate differential pressure sensors is likely to increase.
Leading Manufacturers and Suppliers
A pressure gauge is a method of measuring fluid, gas, water,
or steam intensity in a pressure powered machine to ensure
there are no leaks or pressure changes that would affect the
performance of the system. Pressure systems are designed to
operate within a specific pressure range...
A pressure transducer is a mechanical device that converts
applied pressure, a physical quantity, into a measurable and
industry-standard electrical signal which is linearly and
proportionally related to the applied pressure...